Full waveform digital sonic tool (OYO) has been developed for the purpose of accurately measuring geophysical anomalies in the rockbed containing cracks, and its performance is evaluated by comparing its measurements with those obtained by the conventional sonic logging device (DBM). Modification involves the following. While gain is fixed in the DBM, it is variable in a times10-times200 range in the OYO. Analog transfer:ground surface A/D in the DBM is replaced by digital transfer:intra-probe A/D in the OYO. In the DBM, only a special program running on the MS-DOS can analyze waveform data but, in the OYO, waveforms are recorded in the SEG-Y format enabling the import of the data into generally available waveform processing software. In the OYO, a high-speed communication board is incorporated into the probe, which realizes high-speed communication. There is a very excellent agreement between the two in P-wave velocity distribution as reckoned from the initial run. Regarding the OYO, however, it is pointed out that gain control be performed with the greatest care to prevent waveforms from distortion. 5 figs.