For X-ray image intensifier television operation very low entrance dose rates (about 5.2 nA/kg) are stated and demanded, respectively. These required values are often manifold exceeded in practice so that a check seems to be necessary. It is shown and proved how these measurements can be performed with simple, generally available means of measurement in the radiological practice. For ZnCdS-image intensifiers should be considered that about 13 nA/kg for the large entrance size are not to be exceeded; for the CsI type lower values (factor 1.5) are practicable because of the twofold quantum absorption efficiency. Furthermore, some tests for a semiquantitative function check of the automatic dose rate control are proposed.