## Abstract

The dose speed was measured absorbed for depth zero, (superficial) in soft equivalent fabric, for the secondary pattern{sup s} four sources of beta radiation, (Nr. 86): {sup 90}Sr/{sup 90}Y, (1850 MBq and 74 MBq respectively); {sup 147}Pm, (518 MBq) and {sup 204}TI, (18.5 MBq). The measurement is carried out to different distances of source-detecting separation, (11.0, 30.0 and 50.0 cm for the source of 1850 MBq, 30.0 cm for that of 74 MBq; 11.00 cm for the source of {sup 147}Pmand to contact for all the sources); maintaining the radiation sheaf aligned the one axis of symmetry of the detector, ({alpha} 0 degrees). The detector employed was a extrapolation chambers of variable electrodes and electrode fixed collector, (30 mm of diameter). In accordance with the principle of Bragg-Gray the volume of the chambers is varied and they register the variations of the current of collected ionization, correcting until for a maximum of thirteen correction factors that take into account the deviation to the suppositions that it establishes this principle. The certain values of the speed of superficial absorbed dose are in the following intervals: {sup 90}Sr/{sup 90}Y, (1850 MBq, 0.0, 11.0, 30.0 and 50.0 cm): 43.164 mGy S-t, 0.544 mGy
More>>

## Citation Formats

Alvarez R, J T.
Measurement of dose speed absorbed in depth imparted by sources external secondary patterns of beta radiation. Part 1 Measurement of dose speed absorbed in the surface of soft fabric for isotopes of {sup 90}Sr/{sup 90}Y, {sup 147}Pm and {sup 204}TI; Medicion de rapidez de dosis absorbida en profundidad impartida por fuentes patrones secundarios de radiacion beta externos. Parte 1. Medicion de rapidez de dosis absorbida en la superficie de tejido blando para isotopos de {sup 90}Sr/{sup 90}Y, {sup 147}Pm y {sup 204}TI.
Mexico: N. p.,
1993.
Web.

Alvarez R, J T.
Measurement of dose speed absorbed in depth imparted by sources external secondary patterns of beta radiation. Part 1 Measurement of dose speed absorbed in the surface of soft fabric for isotopes of {sup 90}Sr/{sup 90}Y, {sup 147}Pm and {sup 204}TI; Medicion de rapidez de dosis absorbida en profundidad impartida por fuentes patrones secundarios de radiacion beta externos. Parte 1. Medicion de rapidez de dosis absorbida en la superficie de tejido blando para isotopos de {sup 90}Sr/{sup 90}Y, {sup 147}Pm y {sup 204}TI.
Mexico.

Alvarez R, J T.
1993.
"Measurement of dose speed absorbed in depth imparted by sources external secondary patterns of beta radiation. Part 1 Measurement of dose speed absorbed in the surface of soft fabric for isotopes of {sup 90}Sr/{sup 90}Y, {sup 147}Pm and {sup 204}TI; Medicion de rapidez de dosis absorbida en profundidad impartida por fuentes patrones secundarios de radiacion beta externos. Parte 1. Medicion de rapidez de dosis absorbida en la superficie de tejido blando para isotopos de {sup 90}Sr/{sup 90}Y, {sup 147}Pm y {sup 204}TI."
Mexico.

@misc{etde_20982913,

title = {Measurement of dose speed absorbed in depth imparted by sources external secondary patterns of beta radiation. Part 1 Measurement of dose speed absorbed in the surface of soft fabric for isotopes of {sup 90}Sr/{sup 90}Y, {sup 147}Pm and {sup 204}TI; Medicion de rapidez de dosis absorbida en profundidad impartida por fuentes patrones secundarios de radiacion beta externos. Parte 1. Medicion de rapidez de dosis absorbida en la superficie de tejido blando para isotopos de {sup 90}Sr/{sup 90}Y, {sup 147}Pm y {sup 204}TI}

author = {Alvarez R, J T}

abstractNote = {The dose speed was measured absorbed for depth zero, (superficial) in soft equivalent fabric, for the secondary pattern{sup s} four sources of beta radiation, (Nr. 86): {sup 90}Sr/{sup 90}Y, (1850 MBq and 74 MBq respectively); {sup 147}Pm, (518 MBq) and {sup 204}TI, (18.5 MBq). The measurement is carried out to different distances of source-detecting separation, (11.0, 30.0 and 50.0 cm for the source of 1850 MBq, 30.0 cm for that of 74 MBq; 11.00 cm for the source of {sup 147}Pmand to contact for all the sources); maintaining the radiation sheaf aligned the one axis of symmetry of the detector, ({alpha} 0 degrees). The detector employed was a extrapolation chambers of variable electrodes and electrode fixed collector, (30 mm of diameter). In accordance with the principle of Bragg-Gray the volume of the chambers is varied and they register the variations of the current of collected ionization, correcting until for a maximum of thirteen correction factors that take into account the deviation to the suppositions that it establishes this principle. The certain values of the speed of superficial absorbed dose are in the following intervals: {sup 90}Sr/{sup 90}Y, (1850 MBq, 0.0, 11.0, 30.0 and 50.0 cm): 43.164 mGy S-t, 0.544 mGy s-1 ,0.075 mGy s{sup -1} and 0.027 mGy s{sup -1}, respectively, with a Global Analysis of the order of 1.17%, 1.17%, 1.14% and 1.66%, K J; {sup 90}Sr / {sup 90}Y, (74 MBq, 0.0 and 30 cm): 1.536 mGy s{sup -1} and 0.002 mGy s{sup -1}, with Global Analysis of 1.19.0% and 5.22%, (K = 1) respectively, for the {sup 147}Pm, (0.0 and 11.0 in the interval of: 0.36 {mu}Gy s{sup -1} and 0.43 {mu}Gy s{sup -1}, with one Global Analysis of 1 .42% and 4.28%, (K = 1), respectively; and finally for the {sup 204}TI, (0.0 cm) in the interval of 0.10 {mu}Gy s{sup -1} with a Global Analysis of 1.27%. He calculates of the Global Analysis one carries out of agreement with those recommendations of the BIPM. In all the cases of source-detecting arrangement with separations different from zero, models of simple lineal regression were used. However for the case of the determination of the speed of superficial dose to contact, (it distances source-detecting zero), quadratic regression models were used for the evaluation of the slope of the extrapolation curve, concluding that this model quadratic taking in bill due additional corrections to the deviation of the law of the inverse one with the square of the distance and for effect of 'Shielding' in the collection of electrons in the collector electrode, for this source-detecting type of geometries. Additionally one carries out an indirect mensuration, (through measures of capacity), of the separation of the electrodes for verify it in relation to the position of the zero of the micrometric screw, likewise and of the effective area of the one collector electrode, obtaining the following results: Finally, you recalculation all the values of dose velocity that have been determined for the sources of the PSB in the period from the 15/01/88 to the 31/12/91, since were two systematic errors, which were: an incorrect sequence in the application of the correction factors and a value but exact of {omega}/e 33.97 J/C. Besides considering in these the more precise evaluation of A{sub eff} (Author)}

place = {Mexico}

year = {1993}

month = {Jan}

}

title = {Measurement of dose speed absorbed in depth imparted by sources external secondary patterns of beta radiation. Part 1 Measurement of dose speed absorbed in the surface of soft fabric for isotopes of {sup 90}Sr/{sup 90}Y, {sup 147}Pm and {sup 204}TI; Medicion de rapidez de dosis absorbida en profundidad impartida por fuentes patrones secundarios de radiacion beta externos. Parte 1. Medicion de rapidez de dosis absorbida en la superficie de tejido blando para isotopos de {sup 90}Sr/{sup 90}Y, {sup 147}Pm y {sup 204}TI}

author = {Alvarez R, J T}

abstractNote = {The dose speed was measured absorbed for depth zero, (superficial) in soft equivalent fabric, for the secondary pattern{sup s} four sources of beta radiation, (Nr. 86): {sup 90}Sr/{sup 90}Y, (1850 MBq and 74 MBq respectively); {sup 147}Pm, (518 MBq) and {sup 204}TI, (18.5 MBq). The measurement is carried out to different distances of source-detecting separation, (11.0, 30.0 and 50.0 cm for the source of 1850 MBq, 30.0 cm for that of 74 MBq; 11.00 cm for the source of {sup 147}Pmand to contact for all the sources); maintaining the radiation sheaf aligned the one axis of symmetry of the detector, ({alpha} 0 degrees). The detector employed was a extrapolation chambers of variable electrodes and electrode fixed collector, (30 mm of diameter). In accordance with the principle of Bragg-Gray the volume of the chambers is varied and they register the variations of the current of collected ionization, correcting until for a maximum of thirteen correction factors that take into account the deviation to the suppositions that it establishes this principle. The certain values of the speed of superficial absorbed dose are in the following intervals: {sup 90}Sr/{sup 90}Y, (1850 MBq, 0.0, 11.0, 30.0 and 50.0 cm): 43.164 mGy S-t, 0.544 mGy s-1 ,0.075 mGy s{sup -1} and 0.027 mGy s{sup -1}, respectively, with a Global Analysis of the order of 1.17%, 1.17%, 1.14% and 1.66%, K J; {sup 90}Sr / {sup 90}Y, (74 MBq, 0.0 and 30 cm): 1.536 mGy s{sup -1} and 0.002 mGy s{sup -1}, with Global Analysis of 1.19.0% and 5.22%, (K = 1) respectively, for the {sup 147}Pm, (0.0 and 11.0 in the interval of: 0.36 {mu}Gy s{sup -1} and 0.43 {mu}Gy s{sup -1}, with one Global Analysis of 1 .42% and 4.28%, (K = 1), respectively; and finally for the {sup 204}TI, (0.0 cm) in the interval of 0.10 {mu}Gy s{sup -1} with a Global Analysis of 1.27%. He calculates of the Global Analysis one carries out of agreement with those recommendations of the BIPM. In all the cases of source-detecting arrangement with separations different from zero, models of simple lineal regression were used. However for the case of the determination of the speed of superficial dose to contact, (it distances source-detecting zero), quadratic regression models were used for the evaluation of the slope of the extrapolation curve, concluding that this model quadratic taking in bill due additional corrections to the deviation of the law of the inverse one with the square of the distance and for effect of 'Shielding' in the collection of electrons in the collector electrode, for this source-detecting type of geometries. Additionally one carries out an indirect mensuration, (through measures of capacity), of the separation of the electrodes for verify it in relation to the position of the zero of the micrometric screw, likewise and of the effective area of the one collector electrode, obtaining the following results: Finally, you recalculation all the values of dose velocity that have been determined for the sources of the PSB in the period from the 15/01/88 to the 31/12/91, since were two systematic errors, which were: an incorrect sequence in the application of the correction factors and a value but exact of {omega}/e 33.97 J/C. Besides considering in these the more precise evaluation of A{sub eff} (Author)}

place = {Mexico}

year = {1993}

month = {Jan}

}