The NAA method is the analysis method with high precision and accuracy level. The precision and accuracy of NAA laboratory in P2TRR had been evaluated with SRM 1573a Tomato Leaf and CRM No.9 Sargasso. Samples were irradiated in RSG-GAS's rabbit system and counted with gamma spectrometry. The results obtained from this research are for the SRM 1573a sample the test could analyze 17 elements of the 40 certified elements, and for CRM No.9 sample the test could analyze 16 elements of the 30 certified elements. The result of quantitative analysis indicate the different deviation of each element between 2 -10.5 % compared to the certificate data. This deviation value <15%, this value shows that the precision and accuracy level is good enough.