SUFT78: Minimum Data Set of Measurements for TG 71 Based Electron MonitorUnit Calculations
Abstract
Purpose: Building up a TG71 based electron monitorunit (MU) calculation protocol usually involves massive measurements. This work investigates a minimum data set of measurements and its calculation accuracy and measurement time. Methods: For 6, 9, 12, 16, and 20 MeV of our Varian ClinacSeries linear accelerators, the complete measurements were performed at different depth using 5 square applicators (6, 10, 15, 20 and 25 cm) with different cutouts (2, 3, 4, 6, 10, 15 and 20 cm up to applicator size) for 5 different SSD’s. For each energy, there were 8 PDD scans and 150 point measurements for applicator factors, cutout factors and effective SSDs that were then converted to airgap factors for SSD 99–110cm. The dependence of each dosimetric quantity on field size and SSD was examined to determine the minimum data set of measurements as a subset of the complete measurements. The “missing” data excluded in the minimum data set were approximated by linear or polynomial fitting functions based on the included data. The total measurement time and the calculated electron MU using the minimum and the complete data sets were compared. Results: The minimum data set includes 4 or 5 PDD’s and 51 to 66 point measurementsmore »
 Authors:
 University of Maryland School of Medicine, Baltimore, MD (United States)
 Publication Date:
 OSTI Identifier:
 22642326
 Resource Type:
 Journal Article
 Resource Relation:
 Journal Name: Medical Physics; Journal Volume: 43; Journal Issue: 6; Other Information: (c) 2016 American Association of Physicists in Medicine; Country of input: International Atomic Energy Agency (IAEA)
 Country of Publication:
 United States
 Language:
 English
 Subject:
 61 RADIATION PROTECTION AND DOSIMETRY; ACCURACY; APPROXIMATIONS; DEPTH; LINEAR ACCELERATORS; POLYNOMIALS; RADIATION MONITORING; RADIATION MONITORS
Citation Formats
Xu, H, Guerrero, M, Prado, K, and Yi, B. SUFT78: Minimum Data Set of Measurements for TG 71 Based Electron MonitorUnit Calculations. United States: N. p., 2016.
Web. doi:10.1118/1.4956214.
Xu, H, Guerrero, M, Prado, K, & Yi, B. SUFT78: Minimum Data Set of Measurements for TG 71 Based Electron MonitorUnit Calculations. United States. doi:10.1118/1.4956214.
Xu, H, Guerrero, M, Prado, K, and Yi, B. 2016.
"SUFT78: Minimum Data Set of Measurements for TG 71 Based Electron MonitorUnit Calculations". United States.
doi:10.1118/1.4956214.
@article{osti_22642326,
title = {SUFT78: Minimum Data Set of Measurements for TG 71 Based Electron MonitorUnit Calculations},
author = {Xu, H and Guerrero, M and Prado, K and Yi, B},
abstractNote = {Purpose: Building up a TG71 based electron monitorunit (MU) calculation protocol usually involves massive measurements. This work investigates a minimum data set of measurements and its calculation accuracy and measurement time. Methods: For 6, 9, 12, 16, and 20 MeV of our Varian ClinacSeries linear accelerators, the complete measurements were performed at different depth using 5 square applicators (6, 10, 15, 20 and 25 cm) with different cutouts (2, 3, 4, 6, 10, 15 and 20 cm up to applicator size) for 5 different SSD’s. For each energy, there were 8 PDD scans and 150 point measurements for applicator factors, cutout factors and effective SSDs that were then converted to airgap factors for SSD 99–110cm. The dependence of each dosimetric quantity on field size and SSD was examined to determine the minimum data set of measurements as a subset of the complete measurements. The “missing” data excluded in the minimum data set were approximated by linear or polynomial fitting functions based on the included data. The total measurement time and the calculated electron MU using the minimum and the complete data sets were compared. Results: The minimum data set includes 4 or 5 PDD’s and 51 to 66 point measurements for each electron energy, and more PDD’s and fewer point measurements are generally needed as energy increases. Using only <50% of complete measurement time, the minimum data set generates acceptable MU calculation results compared to those with the complete data set. The PDD difference is within 1 mm and the calculated MU difference is less than 1.5%. Conclusion: Data set measurement for TG71 electron MU calculations can be minimized based on the knowledge of how each dosimetric quantity depends on various setup parameters. The suggested minimum data set allows acceptable MU calculation accuracy and shortens measurement time by a few hours.},
doi = {10.1118/1.4956214},
journal = {Medical Physics},
number = 6,
volume = 43,
place = {United States},
year = 2016,
month = 6
}

A protocol is presented for the calculation of monitor units (MU) for photon and electron beams, delivered with and without beam modifiers, for constant sourcesurface distance (SSD) and sourceaxis distance (SAD) setups. This protocol was written by Task Group 71 of the Therapy Physics Committee of the American Association of Physicists in Medicine (AAPM) and has been formally approved by the AAPM for clinical use. The protocol defines the nomenclature for the dosimetric quantities used in these calculations, along with instructions for their determination and measurement. Calculations are made using the dose per MU under normalization conditions, D{sub 0}{sup ′},more »

Determination of monitor unit check tolerances based on a comparison with measurement and treatment planning system data
ABSTRACT: This work describes the experimental validation of treatment planning system monitor unit (MU) calculations against measurement for a range of scenarios. This, together with a comparison of treatment planning system MUs and an independent MU check method, allows the derivation of confidence intervals for the check process. Data were collected for open and 60° motorized wedge fields using an Elekta Synergy linac at 6 and 8 MV using homogeneous and heterogeneous phantoms. Masterplan (Version 4.0) pencilbeam and collapsed cone algorithms were used for the primary MU calculations with full inhomogeneity correction. Results show that both algorithms agree with measurementmore » 
SUET552: Minimum Monitor Unit Effects On Plan Quality for MultiField Optimized Spot Scanning Proton Therapy
Purpose: To investigate the influence of the minimum monitor unit (MU) on the quality of clinical treatment plans for scanned proton therapy. Methods: Delivery system characteristics limit the minimum number of protons that can be delivered per spot, resulting in a minMU limit. Plan quality can be impacted by the minMU limit. Two sites were used to investigate the impact of minMU on treatment plans: pediatric brain tumor at a depth of 510 cm; a head and neck tumor at a depth of 120 cm. Three field intensity modulated spot scanning proton plans were created for each site with themore » 
Effects of minimum monitor unit threshold on spot scanning proton plan quality
Purpose: To investigate the influence of the minimum monitor unit (MU) on the quality of clinical treatment plans for scanned proton therapy. Methods: Delivery system characteristics limit the minimum number of protons that can be delivered per spot, resulting in a minMU limit. Plan quality can be impacted by the minMU limit. Two sites were used to investigate the impact of minMU on treatment plans: pediatric brain tumor at a depth of 5–10 cm; a head and neck tumor at a depth of 1–20 cm. Threefield, intensity modulated spot scanning proton plans were created for each site with the followingmore »