SUC20106: Small Field Correction Factors for the MicroDiamond Detector in the Gamma KnifeModel C Derived Using Monte Carlo Methods
Abstract
Purpose: To determine small field correction factors for PTW’s microDiamond detector in Elekta’s Gamma Knife ModelC unit. These factors allow the microDiamond to be used in QA measurements of output factors in the Gamma Knife ModelC; additionally, the results also contribute to the discussion on the water equivalence of the relativelynew microDiamond detector and its overall effectiveness in small field applications. Methods: The small field correction factors were calculated as k correction factors according to the Alfonso formalism. An MC model of the Gamma Knife and microDiamond was built with the EGSnrc code system, using BEAMnrc and DOSRZnrc user codes. Validation of the model was accomplished by simulating field output factors and measurement ratios for an available ABS plastic phantom and then comparing simulated results to film measurements, detector measurements, and treatment planning system (TPS) data. Once validated, the final k factors were determined by applying the model to a more waterlike solid water phantom. Results: During validation, all MC methods agreed with experiment within the stated uncertainties: MC determined field output factors agreed within 0.6% of the TPS and 1.4% of film; and MC simulated measurement ratios matched physically measured ratios within 1%. The final k correction factors formore »
 Authors:
 Wayne State University, Detroit, MI (United States)
 (United States)
 Publication Date:
 OSTI Identifier:
 22624311
 Resource Type:
 Journal Article
 Resource Relation:
 Journal Name: Medical Physics; Journal Volume: 43; Journal Issue: 6; Other Information: (c) 2016 American Association of Physicists in Medicine; Country of input: International Atomic Energy Agency (IAEA)
 Country of Publication:
 United States
 Language:
 English
 Subject:
 07 ISOTOPES AND RADIATION SOURCES; 60 APPLIED LIFE SCIENCES; CORRECTIONS; DOSEMETERS; FILMS; MONTE CARLO METHOD; PHANTOMS; VALIDATION
Citation Formats
Barrett, J C, Karmanos Cancer Institute McLarenMacomb, Clinton Township, MI, Knill, C, and Beaumont Hospital, Canton, MI. SUC20106: Small Field Correction Factors for the MicroDiamond Detector in the Gamma KnifeModel C Derived Using Monte Carlo Methods. United States: N. p., 2016.
Web. doi:10.1118/1.4955546.
Barrett, J C, Karmanos Cancer Institute McLarenMacomb, Clinton Township, MI, Knill, C, & Beaumont Hospital, Canton, MI. SUC20106: Small Field Correction Factors for the MicroDiamond Detector in the Gamma KnifeModel C Derived Using Monte Carlo Methods. United States. doi:10.1118/1.4955546.
Barrett, J C, Karmanos Cancer Institute McLarenMacomb, Clinton Township, MI, Knill, C, and Beaumont Hospital, Canton, MI. 2016.
"SUC20106: Small Field Correction Factors for the MicroDiamond Detector in the Gamma KnifeModel C Derived Using Monte Carlo Methods". United States.
doi:10.1118/1.4955546.
@article{osti_22624311,
title = {SUC20106: Small Field Correction Factors for the MicroDiamond Detector in the Gamma KnifeModel C Derived Using Monte Carlo Methods},
author = {Barrett, J C and Karmanos Cancer Institute McLarenMacomb, Clinton Township, MI and Knill, C and Beaumont Hospital, Canton, MI},
abstractNote = {Purpose: To determine small field correction factors for PTW’s microDiamond detector in Elekta’s Gamma Knife ModelC unit. These factors allow the microDiamond to be used in QA measurements of output factors in the Gamma Knife ModelC; additionally, the results also contribute to the discussion on the water equivalence of the relativelynew microDiamond detector and its overall effectiveness in small field applications. Methods: The small field correction factors were calculated as k correction factors according to the Alfonso formalism. An MC model of the Gamma Knife and microDiamond was built with the EGSnrc code system, using BEAMnrc and DOSRZnrc user codes. Validation of the model was accomplished by simulating field output factors and measurement ratios for an available ABS plastic phantom and then comparing simulated results to film measurements, detector measurements, and treatment planning system (TPS) data. Once validated, the final k factors were determined by applying the model to a more waterlike solid water phantom. Results: During validation, all MC methods agreed with experiment within the stated uncertainties: MC determined field output factors agreed within 0.6% of the TPS and 1.4% of film; and MC simulated measurement ratios matched physically measured ratios within 1%. The final k correction factors for the PTW microDiamond in the solid water phantom approached unity to within 0.4%±1.7% for all the helmet sizes except the 4 mm; the 4 mm helmet size overresponded by 3.2%±1.7%, resulting in a k factor of 0.969. Conclusion: Similar to what has been found in the Gamma Knife Perfexion, the PTW microDiamond requires little to no corrections except for the smallest 4 mm field. The overresponse can be corrected via the Alfonso formalism using the correction factors determined in this work. Using the MC calculated correction factors, the PTW microDiamond detector is an effective dosimeter in all available helmet sizes. The authors would like to thank PTW (Friedberg, Germany) for providing the PTW microDiamond detector for this research.},
doi = {10.1118/1.4955546},
journal = {Medical Physics},
number = 6,
volume = 43,
place = {United States},
year = 2016,
month = 6
}

Purpose: Output factor determination for small fields (less than 20 mm) presents significant challenges due to ion chamber volume averaging and diode overresponse. Measured output factor values between detectors are known to have large deviations as field sizes are decreased. No set standard to resolve this difference in measurement exists. We observed differences between measured output factors of up to 14% using two different detectors. Published Monte Carlo derived correction factors were used to address this challenge and decrease the output factor deviation between detectors. Methods: Output factors for Elekta's linacbased stereotactic cone system were measured using the EDGE detectormore »

Determination of the k{sub Q{sub c{sub l{sub i{sub n,Q{sub m{sub s{sub r}{sup f{sub c}{sub l}{sub i}{sub n},f{sub m}{sub s}{sub r}}}}}}}}} correction factors for detectors used with an 800 MU/min CyberKnife{sup ®} system equipped with fixed collimators and a study of detector response to small photon beams using a Monte Carlo method
Purpose: In a previous work, output ratio (OR{sub det}) measurements were performed for the 800 MU/min CyberKnife{sup ®} at the Oscar Lambret Center (COL, France) using several commercially available detectors as well as using two passive dosimeters (EBT2 radiochromic film and microLiF TLD700). The primary aim of the present work was to determine by Monte Carlo calculations the output factor in water (OF{sub MC,w}) and the k{sub Q{sub c{sub l{sub i{sub n,Q{sub m{sub s{sub r}{sup f{sub c}{sub l}{sub i}{sub n},f{sub m}{sub s}{sub r}}}}}}}}} correction factors. The secondary aim was to study the detector response in small beams using Monte Carlomore » 
Output correction factors for nine small field detectors in 6 MV radiation therapy photon beams: A PENELOPE Monte Carlo study
Purpose: To determine detectorspecific output correction factors,k{sub Q} {sub c{sub l{sub i{sub n}}}} {sub ,Q} {sub m{sub s{sub r}}} {sup f{sub {sup {sub c}{sub l}{sub i}{sub n}{sub {sup ,f{sub {sup {sub m}{sub s}{sub r}{sub ,}}}}}}}} in 6 MV small photon beams for air and liquid ionization chambers, silicon diodes, and diamond detectors from two manufacturers. Methods: Field output factors, defined according to the international formalism published byAlfonso et al. [Med. Phys. 35, 5179–5186 (2008)], relate the dosimetry of small photon beams to that of the machinespecific reference field; they include a correction to measured ratios of detector readings, conventionally usedmore » 
A comparison of gammaray buildup factors for lowZ material and for low energies using discrete ordinates and point Monte Carlo methods
This paper discusses the calculation of exposure and absorbed dose buildup factors for a photon point source in infinite beryllium in the lowenergy range of 0.03 to 0.3 MeV, for penetration depth up to 40 mfp, using two discrete ordinates codes, PALLASPL, SPBr, and ANISN. Comparisons of both result to values obtained by point Monte Carlo calculations using the electron gamma shower version 4 code show reasonable agreement for two types of sources: normally incident and point isotropic sources. The fitting parameters of a geometricprogression method formula are determined for the resulting buildup factor data. These fitting parameters are inmore » 
Study of scattered photons from the collimator system of Leksell Gamma Knife using the EGS4 Monte Carlo Code
In the algorithm of Leksell GAMMAPLAN (the treatment planning software of Leksell Gamma Knife), scattered photons from the collimator system are presumed to have negligible effects on the Gamma Knife dosimetry. In this study, we used the EGS4 Monte Carlo (MC) technique to study the scattered photons coming out of the single beam channel of Leksell Gamma Knife. The PRESTA (Parameter Reduced ElectronStep Transport Algorithm) version of the EGS4 (Electron Gamma Shower version 4) MC computer code was employed. We simulated the single beam channel of Leksell Gamma Knife with the full geometry. Primary photons were sampled from within themore »