Precise and direct method for the measurement of the torsion spring constant of the atomic force microscopy cantilevers
Abstract
A direct method for the evaluation of the torsional spring constants of the atomic force microscope cantilevers is presented in this paper. The method uses a nanoindenter to apply forces at the long axis of the cantilever and in the certain distance from it. The torque vs torsion relation is then evaluated by the comparison of the results of the indentations experiments at different positions on the cantilever. Next, this relation is used for the precise determination of the torsional spring constant of the cantilever. The statistical analysis shows that the standard deviation of the calibration measurements is equal to approximately 1%. Furthermore, a simple method for calibration of the photodetector’s lateral response is proposed. The overall procedure of the lateral calibration constant determination has the accuracy approximately equal to 10%.
 Authors:
 Institute of Fundamental Technological Research, Pawińskiego 5b, 02106 Warsaw (Poland)
 Publication Date:
 OSTI Identifier:
 22392315
 Resource Type:
 Journal Article
 Resource Relation:
 Journal Name: Review of Scientific Instruments; Journal Volume: 86; Journal Issue: 1; Other Information: (c) 2014 AIP Publishing LLC; Country of input: International Atomic Energy Agency (IAEA)
 Country of Publication:
 United States
 Language:
 English
 Subject:
 46 INSTRUMENTATION RELATED TO NUCLEAR SCIENCE AND TECHNOLOGY; ACCURACY; CALIBRATION; COMPARATIVE EVALUATIONS; DISTANCE; MICROSCOPES; MICROSCOPY; SPRINGS; TORQUE; TORSION
Citation Formats
Jarząbek, D. M., Email: djarz@ippt.pan.pl. Precise and direct method for the measurement of the torsion spring constant of the atomic force microscopy cantilevers. United States: N. p., 2015.
Web. doi:10.1063/1.4904866.
Jarząbek, D. M., Email: djarz@ippt.pan.pl. Precise and direct method for the measurement of the torsion spring constant of the atomic force microscopy cantilevers. United States. doi:10.1063/1.4904866.
Jarząbek, D. M., Email: djarz@ippt.pan.pl. 2015.
"Precise and direct method for the measurement of the torsion spring constant of the atomic force microscopy cantilevers". United States.
doi:10.1063/1.4904866.
@article{osti_22392315,
title = {Precise and direct method for the measurement of the torsion spring constant of the atomic force microscopy cantilevers},
author = {Jarząbek, D. M., Email: djarz@ippt.pan.pl},
abstractNote = {A direct method for the evaluation of the torsional spring constants of the atomic force microscope cantilevers is presented in this paper. The method uses a nanoindenter to apply forces at the long axis of the cantilever and in the certain distance from it. The torque vs torsion relation is then evaluated by the comparison of the results of the indentations experiments at different positions on the cantilever. Next, this relation is used for the precise determination of the torsional spring constant of the cantilever. The statistical analysis shows that the standard deviation of the calibration measurements is equal to approximately 1%. Furthermore, a simple method for calibration of the photodetector’s lateral response is proposed. The overall procedure of the lateral calibration constant determination has the accuracy approximately equal to 10%.},
doi = {10.1063/1.4904866},
journal = {Review of Scientific Instruments},
number = 1,
volume = 86,
place = {United States},
year = 2015,
month = 1
}

Spring constant calibration of atomic force microscopy cantilevers with a piezosensor transfer standard
We describe a method to calibrate the spring constants of cantilevers for atomic force microscopy (AFM). The method makes use of a ''piezosensor'' composed of a piezoresistive cantilever and accompanying electronics. The piezosensor was calibrated before use with an absolute force standard, the NIST electrostatic force balance (EFB). In this way, the piezosensor acts as a force transfer standard traceable to the International System of Units. Seven singlecrystal silicon cantilevers with rectangular geometries and nominal spring constants from 0.2 to 40 N/m were measured with the piezosensor method. The values obtained for the spring constant were compared to measurements bymore » 
Accurate spring constant calibration for very stiff atomic force microscopy cantilevers
There are many atomic force microscopy (AFM) applications that rely on quantifying the force between the AFM cantilever tip and the sample. The AFM does not explicitly measure force, however, so in such cases knowledge of the cantilever stiffness is required. In most cases, the forces of interest are very small, thus compliant cantilevers are used. A number of methods have been developed that are well suited to measuring low stiffness values. However, in some cases a cantilever with much greater stiffness is required. Thus, a direct, traceable method for calibrating very stiff (approximately 200 N/m) cantilevers is presented here.more » 
Note: Spring constant calibration of nanosurfaceengineered atomic force microscopy cantilevers
The determination of the dynamic spring constant (k{sub d}) of atomic force microscopy cantilevers is of crucial importance for converting cantilever deflection to accurate force data. Indeed, the nondestructive, fast, and accurate measurement method of the cantilever dynamic spring constant by Sader et al. [Rev. Sci. Instrum. 83, 103705 (2012)] is confirmed here for plane geometry but surface modified cantilevers. It is found that the measured spring constants (k{sub eff}, the dynamic one k{sub d}), and the calculated (k{sub d,1}) are in good agreement within less than 10% error. 
Spring constant calibration of atomic force microscope cantilevers of arbitrary shape
The spring constant of an atomic force microscope cantilever is often needed for quantitative measurements. The calibration method of Sader et al. [Rev. Sci. Instrum. 70, 3967 (1999)] for a rectangular cantilever requires measurement of the resonant frequency and quality factor in fluid (typically air), and knowledge of its plan view dimensions. This intrinsically uses the hydrodynamic function for a cantilever of rectangular plan view geometry. Here, we present hydrodynamic functions for a series of irregular and nonrectangular atomic force microscope cantilevers that are commonly used in practice. Cantilever geometries of arrow shape, small aspect ratio rectangular, quasirectangular, irregular rectangular,more »