Instrument Qualification of Custom Fabricated Water Activity Meter for Hot Cell Use
This report describes a custom fabricated water activity meter and the results of the qualification of this meter as described in the laboratory test plan LAB-PLN-11-00012, Testing and Validation of an Enhanced Acquisition and Control System. It was calibrated against several NaOH solutions of varying concentrations to quantify the accuracy and precision of the instrument at 20 °C and 60 °C. Also, a schematic and parts list of the equipment used to make the water activity meter will be presented in this report.
- Publication Date:
- OSTI Identifier:
- Report Number(s):
- LAB-RPT-13-00006 Rev 0
- DOE Contract Number:
- Resource Type:
- Technical Report
- Research Org:
- HNF (Hanford Site (HNF), Richland, WA (United States))
- Sponsoring Org:
- USDOE Office of Environmental Management (EM)
- Contributing Orgs:
- Washington River Protection Solutions LLC.
- Country of Publication:
- United States
- 46 INSTRUMENTATION RELATED TO NUCLEAR SCIENCE AND TECHNOLOGY
Enter terms in the toolbar above to search the full text of this document for pages containing specific keywords.