Summary: GC SOLUTION SOFTWARE
Craig S Young
Sr. Technical Support
Shimadzu Scientific Instruments
GC SOLUTION Two Modules
GC Real Time Analysis & GC PostRun
Use Real Time Analysis to set the instrument configuration.
Use Real Time Analysis to set run methods and acquire data.
Use PostRun to process data, integrate peaks, and perform quantitation.
Real Time Analysis
Initialize the instrument
Turn on your GC and establish gas flow for the carrier and any detector gases. Refer to the end
of this document for the information about the gases to be used for GC. Open the GC Solution
Real Time Analysis module. From the View menu, choose Instrument Monitor. The Monitor
appears on the right hand margin of the screen. Click on System Start. Also click ON any
modules that are currently OFF (detector, injector, column oven). The GC will attempt to reach
temperature and flow set points for any devices on all analytical lines according to the method
last used. Analytical lines (separate physical analysis paths made up of at least one injector and
one detector) are distinguished by separate tabs on this Instrument Monitor screen. The